Validity and practical utility of accelerometry for the measurement of in-hand physical activity in horses
© Morrison et al. 2015
Received: 26 February 2014
Accepted: 1 September 2015
Published: 11 September 2015
Accelerometers are valid, practical and reliable tools for the measurement of habitual physical activity (PA). Quantification of PA in horses is desirable for use in research and clinical settings. The objective of this study was to evaluate a triaxial accelerometer for objective measurement of PA in the horse by assessment of their practical utility and validity.
Horses were recruited to establish both the optimal site of accelerometer attachment and questionnaire designed to explore owner acceptance. Validity and cut-off values were obtained by assessing PA at various gaits. Validation study- 20 horses wore the accelerometer while being filmed for 10 min each of rest, walking and trotting and 5 mins of canter work. Practical utility study- five horses wore accelerometers on polls and withers for 18 h; compliance and relative data losses were quantified.
Accelerometry output differed significantly between the four PA levels (P < 0•001) for both wither and poll placement. For withers placement, ROC analyses found optimal sensitivity and specificity at a cut-off of <47 counts per minute (cpm) for rest (sensitivity 99.5 %, specificity 100 %), 967–2424 cpm for trotting (sensitivity 96.7 %, specificity 100 %) and ≥2425 cpm for cantering (sensitivity 96.0 %, specificity 97.0 %). Attachment at the poll resulted in optimal sensitivity and specificity at a cut-off of <707 counts per minute (cpm) for rest (sensitivity 97.5 %, specificity 99.6 %), 1546–2609 cpm for trotting (sensitivity 90.33 %, specificity 79.25 %) and ≥2610 cpm for cantering (sensitivity 100 %, specificity 100 %) In terms of practical utility, accelerometry was well tolerated and owner acceptance high.
Accelerometry data correlated well with varying levels of in-hand equine activity. The use of accelerometers is a valid method for objective measurement of controlled PA in the horse.
KeywordsHorse Physical activity Accelerometry Obesity Validation
Accelerometers are small, non-invasive, portable devices that can be used to quantify the intensity, duration and frequency of physical activity (PA). These devices have been used extensively over the past decade in humans, both adults and children, and more recently in animals including dogs [1–4] cats  and cows  to measure the amount and frequency of movement. However, few studies have reported using accelerometers for the measurement of PA in horses .
In humans, accelerometers are typically worn on the hip, close to the body’s centre of mass, and can accurately record PA without incurring recall bias from the subject . Accelerometry is now regarded as a reliable mechanism with which to collect information on free-living habitual PA [9–11]. It has facilitated a number of studies in which PA levels are of interest such as in obesity where accelerometry has been used to improve the knowledge of its aetiology, prevention and treatment [12–17].
In veterinary medicine, accelerometers have been shown to be valid, practical and reliable for the measurement of habitual PA in dogs  and subsequently the technique has been applied for objective evaluation of PA in dogs during a weight loss programme . As the devices are portable, lightweight and non-invasive they could potentially be used in horses for measurement of PA. As in other species, equine obesity has a relatively high prevalence and estimates in UK and Scotland range from 21 to 45 % [19, 20] and predisposes to important conditions such as laminitis and equine metabolic syndrome . The ability to quantitatively and objectively monitor the PA levels of a horse using accelerometers is highly desirable, however as in humans, requires evaluation and validation prior to it being used for research into PA in relation to obesity or many of the other potential applications. The measurement of PA in horses has been validated previously using accelerometer multidirectional accelerometer1  and more recently with a pedometer2 . However, as yet there are no published studies that describe using one of the most commonly used triaxial accelerometers3 in human PA research to measure PA in horses.
The purpose of this investigation was to evaluate a triaxial accelerometer for the objective measurement of PA in the horse by assessment of its practical utility, and validity during controlled PA. Firstly, it was important to establish that the accelerometer was practical to use in the turned out horse and that it is well tolerated by horses and acceptable to owners. For this, the optimal site of attachment of the accelerometer needed to be determined. Secondly, in human and animal studies, accelerometry is usually validated against either directly observed movement or against energy expenditure, both methods being regarded widely as ‘gold standards’ for the validation of movement [2, 9–11]. The objective of the validation was to test that accelerometry output increased significantly as directly observed intensity of in-hand equine PA increased. Support for this hypothesis would thus support the validity of accelerometery for measuring controlled PA in the horse.
Two studies were undertaken–practical utility, and validation. The accelerometer used during these studies is a small, water-resistant, portable device (46 × 33 × 15 mm, weighing 19 g) that records acceleration along three axes (defined as X, Y, Z by the manufaturer). Powered by a lithium ion polymer battery, it is capable of recording accelerations in the range ±6 g (gravitational acceleration) for up to 30 days. The GT3-X+ can record data in sampling rates between 30 and 100 Hz, as specified by the user. In this study the accelerometer was set to sample date at 30 Hz. The raw data is then band pass filtered at a bandwidth of 0.25 to 2.5 Hz intended to eliminate high frequency artefact vibrations, and converted to an activity count using a proprietary algorithm. The recorded activity counts are then summed over a period of time specified by the user, known as an epoch. In these investigations the accelerometer recorded the PA of each horse in 15 s epochs, but these were then summarised over 1 min periods to give a mean accelerometer count per minute. Activity data recorded by the accelerometer can be downloaded using the manufacturers accompanying software4 which outputs the activity countfor each individual axis and also a calculated vector magnitude of the three individual axes combined.
Practical utility study
Initially the most suitable site of attachment on the horse and practical utility of the accelerometer were investigated in a convenience sample of five horses. Evaluation of the practical utility of each potential site was considered to include ease of attachment, longevity of attachment and data loss/absence. Cable ties were used to attach the accelerometer to the head collar at the poll of the horse. For the withers, each horse either wore a nylon stretch bib usually worn to protect the shoulders, chest and withers from rug rub, with accelerometers attached via hook and loop tape or attachment of an accelerometer to a surcingle 10 cm below the withers. Duct tape was used to attach accelerometers to the sternum and the sacrum.
Five accelerometers at the aforementioned sites, recording PA in 15 s epochs, were worn simultaneously by each horse for 2 h over different levels of PA. Any problems experienced with the accelerometers over the study period were noted by one of the researchers.
Practical utility of attachment of the accelerometers to the poll and withers (attachment to a head collar and surcingle respectively) was further investigated for 18 consecutive hours in five horses. The percentage of possible minutes lost (18 h gave a potential of 1080 min of accelerometry per horse) was calculated to quantify data loss or absence.
A brief questionnaire with Likert scale responses was designed to evaluate owner acceptance of attaching the accelerometer for a period of 7 days to six possible sites, namely; the poll, withers, sternum, foreleg (above the knee), hind leg (proximal to the metatarsophalangeal joint) and sacrum. Owners were asked to indicate if they would be very willing, willing, indifferent, unwilling or very unwilling to allow attachment at each site.
Sedentary–when the horse was at rest in its stable, with free movement of the head and slight movement of the trunk.
Light intensity physical activity–walking the horse (four-beat gait), with slow translocation of the trunk.
Moderate intensity physical activity–a slow to fast trot (two-beat gait), with moderate translocation of the trunk.
Vigorous intensity physical activity–cantering (three-beat gait), with fast translocation of the trunk.
Sedentary behaviour was measured when horses were rested in their stable without human contact (hay and water was provided ad libitum). Light intensity PA was measured by walking horses in-hand in a sand arena at constant pace. Moderate and vigorous intensities of PA were measured as horses were lunged in a sand arena by an experienced person at constant pace. After successful completion of each stage horses were rested for at least 5 min before moving on to the next stage. The raw 15-s epochs from the accelerometer were extracted for 10 clean minutes (i.e. when a horse performed a single intensity of PA) at activity levels 1 to 3 and for 5 clean minutes at activity level 4 for each horse and then summarised as a count per minute for the vertical and integrated axis.
Accelerometry output was highly skewed, with a large number of ‘zeros’ in the sedentary behaviour category (generally while the horses stood in the stable), and so a non-parametric approach to analysis was taken in the validation study. Differences in median accelerometry output between the categories of PA described above (1–4) were tested for statistical significance using the Friedman Test, with significance level at 5 % in all tests. Where significant differences were observed with the Friedman test, the location of significant differences was tested using Wilcoxon tests between pairs of PA categories 1–4. All analyses in the validation study were conducted using commercially available software.5
Categorisation of intensity of PA
Using the same sample as in the validation study, receiver operating characteristic curve (ROC) analyses were conducted using commercially available software6 to establish accelerometer cut points to categorise activity by intensity (sedentary, light PA, moderate PA and vigorous PA as described above). The analysis looked at each individual cpm in the data set to establish readings with the highest average sensitivity (probability of categorising a horse’s activity level correctly e.g. periods of time when the horse is actually sedentary being recorded as sedentary by the accelerometer) and specificity (probability of incorrectly categorising the activity level, e.g. light intensity PA might be recorded as sedentary). The average accelerometer cpm (both vertical and integrated axes) for each clean minute of activity recorded for all horses was modelled as the independent variable. The dependent variable was calculated by coding PA categories as either 0 or 1 depending on the boundary being generated; i.e. for sedentary behaviour this corresponded to all activities in category 1 being coded as 1 and all non-sedentary activities (categories 2–4) being coded as 0; for moderate PA this corresponded to all activities in category 3 being coded as 1 and all non-trotting activities (categories 1, 2 and 4) being coded as 0; for vigorous PA this corresponded to all activities in category 4 being coded as 1 and all non-cantering activities (categories 1–3) being coded as 0. The sedentary and moderate PA cut points provided the boundaries for the light PA category.
Contingency tables were produced and weighted Kappa (κ) calculated to determine the classification accuracy of the cut points developed in ROC analyses. κ is a measure of inter-rater agreement after agreement due to chance is taken out of the equation , with strength of score rated as; ‘poor’ (<0.20); ‘fair’ (0.21–0.40); ‘moderate’ (0.41–0.60); ‘good’ (0.61–0.8); and ‘very good’ (0.81–1.00) .
The study was approved by the University of Glasgow, School of Veterinary Medicine Ethics and Welfare Committee. Informed written consent to participation was received from each horse owner.
Practical utility study
The convenience sample of five horses recruited to the practical utility study had a mean age of 16.8 years and height range of 1.30 to 1.65 m. Four mare’s and one gelding of varying breeds were included. Attachment of the accelerometers caused no alterations of normal behaviour although some horses proceeded to roll shortly after attachment, but quickly became accustomed to the device. Accelerometers located on the poll showed cpm during sedentary behaviour to be greater than acceleration counts recorded at other locations.
Complete 1080 min data sets were obtained from accelerometers attached to both the withers and poll. However, there were incomplete data sets for accelerometers attached at the foreleg, sternum and sacrum due to their detachment during vigorous PA.
Forty-two owners completed an owner acceptance questionnaire. The sites with the highest proportion of owners willing or very willing to allow an accelerometer to be attached were: poll (29/42 (69 %)); foreleg (27/42 (64 %)); and withers (25/42 (60 %)). The sites with the highest proportion of owners unwilling or very unwilling to allow an accelerometer to be attached were: sternum (20/42 (48 %)); sacrum (17/42 (40 %)); and hind leg (16/42 (38 %)).
In view of the findings of the practical utility assessment and owner questionnaire, the poll and withers were thought to be the most suitable sites for accelerometer attachment for the validity study.
Categorisation of intensity of PA; establishment of cut point determination
Sensitivity, specificity and area under the ROC curve for the development of accelerometer cut points for wither and poll placement
Area under ROC curve (95 % CI)
Cut points (cpm)
Classification accuracy of accelerometer when predicting PA (withers placement; n = 20)
Classification accuracy of accelerometer when predicting PA (poll placement; n = 20)
When the data using the poll placement were analysed, 194/200 (97 %) minutes were correctly classified as sedentary behaviour, with the remaining 6 (3 %) minutes incorrectly classified as light intensity PA. In the light intensity PA category 114/200 (57 %) minutes were correctly classified, 2/200 (1 %) minutes were incorrectly classified as sedentary behavior, 83/200 (42 %) minutes were incorrectly classified as moderate intensity PA, and 1/200 (1 %) minutes was incorrectly classified as vigorous intensity PA. In the moderate intensity PA category 171/200 (85 %) minutes were correctly classified, and 29/200 (15 %) minutes were incorrectly classified as light intensity PA. Finally, in the vigorous intensity PA category, 100/100 (100 %) minutes were correctly classified. Overall 663/700 (95 %) minutes were correctly classified using the withers placement data, and 579/700 (83 %) minutes correctly classified using the poll placement data.
The objective assessment of PA in the horse has many useful applications in both health and disease. In this study validity of an accelerometer based activity monitor was established for sedentary, light, moderate and vigorous PA in the stabled horse and when exercised in-hand and practical utility established. The practical utility study focused on evaluating the best location to attach the accelerometer to a horse. Although all four locations of accelerometer attachment in the practical utility study were successful at recording the intensity and duration of PA, issues arose with maintaining attachment of the devices at the sacrum and sternum and hence data loss occurred. The most acceptable attachment sites were the poll and withers, using a head collar and a surcingle or nylon bib respectively. These locations and the method of attachment were well tolerated by horses and were acceptable to most owners.
While both the poll and withers could be used as sites for attachment of accelerometers for future analysis of equine PA, the withers may be preferred if only a single site of attachment is used. When measuring habitual PA in humans, accelerometers are commonly worn on the trunk, usually at the hip, as this site has been shown to provide a better measure of PA than accelerometers attached to the limbs . Preliminary studies in the dog also evaluated several sites of location for accelerometers and attachment to the collar was considered most convenient and reliable  and is now routinely used [2–4]. Withers placement in the horse is most likely to measure PA in terms of distance travelled and movement, whilst an accelerometer attached to the poll may also measure motion not associated with traveling such as movement of the head whilst eating. This effect of ‘motion without travel’ is one reason suggested for not placing accelerometers on dogs’ limbs when assessing PA  and the same argument could be extrapolated to the horse. Furthermore, the better classification accuracy of the accelerometer attached to the withers (95 % of minutes classified correctly) compared to the poll attachment (83 % of minutes classified correctly) suggests that the withers may be the preferred site of attachment. Accelerometers located on the poll also showed cpm during sedentary behaviour to be greater than acceleration counts recorded at other locations which was thought to reflect movement of the head. However, it is possible that future studies may incorporate attachment at both sites during data collection. This may facilitate discrimination between whole body movements and movement of the head/neck during standing/resting.
In the absence of previous data, the optimal sample size for the validation study was difficult to assess. In human accelerometry validation studies, a sample size of 20 to 30 is typical [9–11]: analysis of a sample of 20 horses was therefore considered adequate for this study. A diverse sample of horses within a height range of 1.57 to 1.78 m was recruited to provide adequate tests of validity and to enhance the applicability of study findings. Further studies would be warranted to validate the use of accelerometry in specific equine types and breed groups should specific information be required about PA in one of these; for instance, cut points for varying PA levels may need to be determined by breed as it is likely, for example, that the threshold for vigorous exercise may be higher for a racing thoroughbred than for a pony. Other differences may also exist for the other categories of PA.
During the validation study there was excellent correlation between vertical and integrated axes suggesting that in future studies one or other of these measurements could be used, particularly if the accelerometer is attached in such a way that slippage becomes unlikely and plane of movement remains unchanged.
In order to measure both total volume and the intensity of PA among horses it is necessary to determine accelerometer cut points that could classify PA by intensity. The ability to categorise PA by intensity may in the future facilitate a clearer understanding of what particular dimensions of PA are most important for equine health. Applications may include assessment of PA in laminitis, obese individuals and geriatrics, or in post-operative horses as an indicator of pain. Due to its quantitative nature, accelerometry could also facilitate accurate assessment of the effects of clinical interventions on cumulative PA. The addition of the accelerometer used in the present study as a tool for evaluating PA also offers greater choice to the equine PA researcher interested in assessing both the total volume and intensity of PA.
There were a number of strengths and weaknesses to this study. The sample size was acceptable for the validation study, and in line with human and canine studies. Another strength was that the study investigated two important aspects of accelerometer use, namely, validation, and practical utility, both of which are important for future use of the accelerometer in equine studies. Finally the study was structured to include a range of activities that should mimic PA when horses are exercised in-hand.
A limitation could be considered to be the smaller sample size for the practical utility study and the short duration of measurement (18 h). Furthermore, the study design did not allow for the assessment of typical horse behavior when turned out in the field. Turned out horses will typically spend much of their time grazing and only move a few steps at a time before stopping again . In addition, although the results presented here show that triaxial accelerometry is capable of classifying horse PA when exercised in-hand it is unclear whether it can discriminate between resting activities such as grazing and true “quiet standing” or sleep. Further studies which aim to assess the validity of accelerometry when discriminating between quiet standing and grazing, as opposed to discriminating between continuous bouts of PA of varying intensity as in the present study, are required. Another limitation of the study is that the position of the accelerometers were monitored throughout data collection which limits the applicability of the findings to studies where this is not possible. Substantial slippage of the device may result in acceleration being recorded at an unintended site and therefore invalidate the results.
It was disappointing when accelerometers became detached during the practical utility study, and this was due to activities such as rolling of the horse. It is important that this limitation of accelerometer use in the horse is recognised as, at the time of designing the study the accelerometer used cost $250 each and are therefore expensive to replace if lost or damaged. However, attention to secure attachment of accelerometers using multiple cable ties, and protection of accelerometers in fabric pockets should be considered to minimise such losses and breakages. During the validation study horses were lunged when performing activities. It is therefore possible that the resulting centripetal accelerations could be measured by the accelerometer in addition to the gravitational acceleration. However, the use of band pass filtered data may remove the centripetal acceleration. Future studies should determine whether this is the case.
The results of this study show that triaxial accelerometry is a practical and valid method for measuring PA of the horse when exercised in-hand and paves the way for future applications of triaxial accelerometry in equine research.
Further studies are required to determine whether triaxial accelerometry is valid when measuring PA in the turned out horse which would facilitate a number of studies in which habitual PA levels are of interest such as in obesity where accelerometry has been used to improve the knowledge of its aetiology, prevention and treatment [12–17, 28].
This study suggests that a triaxial accelerometer is a valid device for the objective measurement of in-hand PA in horses. This technique should facilitate future research where quantitative information regarding controlled equine PA is required, and may therefore be of significant clinical and academic value.
Actiwatch, Cambridge Neurotechnology Ltd, UK
Activity Lying Temperature Pedometer, Ingenieurbuero Holz, Falkenhagen, Germany
ActiGraph GT3X+ accelerometer, ActiGraph, Florida, USA
Actilife, ActiGraph, Florida, USA
Minitab 16.1.1, Minitab Inc. State College, PA.
Medcalc 22.214.171.124, Medcalc Software, Ostend, Belgium.
Funding was provided by the University of Glasgow, School of Veterinary Medicine. Funding was also obtained through BBSRC Research Experience Placement scheme and Wellcome Trust Vacation Scholarships for which we are grateful.
Open AccessThis article is distributed under the terms of the Creative Commons Attribution 4.0 International License (http://creativecommons.org/licenses/by/4.0/), which permits unrestricted use, distribution, and reproduction in any medium, provided you give appropriate credit to the original author(s) and the source, provide a link to the Creative Commons license, and indicate if changes were made. The Creative Commons Public Domain Dedication waiver (http://creativecommons.org/publicdomain/zero/1.0/) applies to the data made available in this article, unless otherwise stated.
- Hansen BD, Lascelles DX, Keene BW, Adams AK, Thomson AE. Evaluation of an accelerometer for at-home monitoring of spontaneous activity in dogs. Am J Vet Res. 2007;68(5):468–75.View ArticlePubMedGoogle Scholar
- Yam PS, Penpraze V, Young D, Todd MS, Cloney AD, Houston-Callaghan KA, et al. Validity, practical utility and reliability of Actigraph accelerometry for the measurement of habitual physical activity in dogs. J Small Anim Pract. 2011;52(2):86–92.View ArticlePubMedGoogle Scholar
- Morrison R, Penpraze V, Beber A, Reilly JJ, Yam PS. Associations between obesity and physical activity in dogs: a preliminary investigation. J Small Anim Pract. 2013;54(11):570–4.View ArticlePubMedGoogle Scholar
- Morrison R, Penpraze V, Greening R, Underwood T, Reilly JJ, Yam PS. Correlates of objectively measured physical activity in dogs. Vet J. 2014;199(2):263–7.View ArticlePubMedGoogle Scholar
- Lascelles BD, Hansen BD, Thomson A, Pierce CC, Boland E, Smith ES. Evaluation of a digitally integrated accelerometer-based activity monitor for the measurement of activity in cats. Vet Anaesth Analg. 2008;35(2):173–83.View ArticlePubMedGoogle Scholar
- Robert B, White BJ, Renter DG, Larson RL. Evaluation of three-dimensional accelerometers to monitor and classify behavior patterns in cattle. Comput Electron Agric. 2009;67(1–2):80–4.View ArticleGoogle Scholar
- Burla J-B, Ostertag A, Schulze Westerath H, Hillmann E. Gait determination and activity measurement in horses using an accelerometer. Comput Electron Agric. 2014;102:127–33.View ArticleGoogle Scholar
- Freedson PS, Melanson E, Sirard J. Calibration of the Computer Science and Applications, Inc. accelerometer. Med Sci Sports Exerc. 1998;30(5):777–81.View ArticlePubMedGoogle Scholar
- Corder K, Ekelund U, Steele RM, Wareham NJ, Brage S. Assessment of physical activity in youth. J Appl Physiol. 2008;105(3):977–87.View ArticlePubMedGoogle Scholar
- Reilly JJ, Penpraze V, Hislop J, Davies G, Grant S, Paton JY. Objective measurement of physical activity and sedentary behaviour: review with new data. Arch Dis Child. 2008;93(7):614–9.View ArticlePubMedGoogle Scholar
- De Vries SI, Van Hirtum HW, Bakker I, Hopman-Rock M, Hirasing RA, Van Mechelen W. Validity and reproducibility of motion sensors in youth: a systematic update. Med Sci Sports Exerc. 2009;41(4):818–27.View ArticlePubMedGoogle Scholar
- Westerterp KR. Pattern and intensity of physical activity. Nature. 2001;410(6828):539.View ArticlePubMedGoogle Scholar
- Montgomery C, Reilly JJ, Jackson DM, Kelly LA, Slater C, Paton JY, et al. Relation between physical activity and energy expenditure in a representative sample of young children. Am J Clin Nutr. 2004;80(3):591–6.PubMedGoogle Scholar
- Reilly JJ, Kelly L, Montgomery C, Williamson A, Fisher A, McColl JH, et al. Physical activity to prevent obesity in young children: cluster randomised controlled trial. Br Med J. 2006;333(7577):1041–3.View ArticleGoogle Scholar
- Ness AR, Leary SD, Mattocks C, Blair SN, Reilly JJ, Wells J, et al. Objectively measured physical activity and fat mass in a large cohort of children. PLoS Med. 2007;4(3):e97.PubMed CentralView ArticlePubMedGoogle Scholar
- Hughes AR, Stewart L, Chapple J, McColl JH, Donaldson MDC, Kelnar CJH, et al. Randomized, controlled trial of a best-practice individualized behavioral program for treatment of childhood overweight: Scottish Childhood Overweight Treatment Trial (SCOTT). Pediatrics. 2008;121(3):e539–46.View ArticlePubMedGoogle Scholar
- Robertson W, Stewart-Brown S, Wilcock E, Oldfield M, Thorogood M. Utility of accelerometers to measure physical activity in children attending an obesity treatment intervention. J Obes. 2011;2011.Google Scholar
- Morrison R, Reilly JJ, Penpraze V, Pendlebury E, Yam PS. A 6-month observational study of changes in objectively measured physical activity during weight loss in dogs. J Small Anim Pract. 2014;55(11):566–70.View ArticlePubMedGoogle Scholar
- Stephenson HM, Green MJ, Freeman SL. Prevalence of obesity in a population of horses in the UK. Vet Rec. 2011;168(5):131.View ArticlePubMedGoogle Scholar
- Wyse CA, McNie KA, Tannahil VJ, Love S, Murray JK. Prevalence of obesity in riding horses in Scotland. Vet Rec. 2008;162(18):590–1.View ArticlePubMedGoogle Scholar
- Treiber KH, Kronfeld DS, Hess TM, Byrd BM, Splan RK, Staniar WB. Evaluation of genetic and metabolic predispositions and nutritional risk factors for pasture-associated laminitis in ponies. J Am Vet Med Assoc. 2006;228(10):1538–45.View ArticlePubMedGoogle Scholar
- Piccione G, Costa A, Giannetto C, Caola G. Daily rhythms of activity in horses housed in different stabling conditions. Biol Rhythm Res. 2007;39(1):79–84.View ArticleGoogle Scholar
- O’Hara NM, Baranowski T, Simonsmorton BG, Wilson BS, Parcel GS. Validity of the observation of childrens physical activity. Res Q Exerc Sport. 1989;60(1):42–7.View ArticlePubMedGoogle Scholar
- Feuerman M, Miller AR. Relationships between statistical measures of agreement: sensitivity, specificity and kappa. J Eval Clin Pract. 2008;14(5):930–3.View ArticlePubMedGoogle Scholar
- Altman DG. Practical statistics for medical research / Douglas G. Altman. London: Chapman & Hall; 1991.Google Scholar
- Bouten CV, Sauren AA, Verduin M, Janssen JD. Effects of placement and orientation of body-fixed accelerometers on the assessment of energy expenditure during walking. Med Biol Eng Comput. 1997;35(1):50–6.View ArticlePubMedGoogle Scholar
- Hampson BA, Morton JM, Mills PC, Trotter MG, Lamb DW, Pollitt CC. Monitoring distances travelled by horses using GPS tracking collars. Aust Vet J. 2010;88(5):176–81.View ArticlePubMedGoogle Scholar
- Mattocks C, Leary S, Ness A, Deere K, Saunders J, Tilling K, et al. Calibration of an accelerometer during free-living activities in children. Int J Pediatr Obes. 2007;2(4):218–26.View ArticlePubMedGoogle Scholar